|Home > Astronomy > Wide Field Camera > Detector Binning and Windowing|
Detector Binning and WindowingEffective 10 August 2010, we can offer binning in all 4 WFC chips and windowing in chip 4 only.
WFC binning can be set with the following command:
SYS > bin 1 1 (1x1 pixels, no binning, by default)
Be aware, binning increases the noise by about 2-5 times, while CCD2 is dominated by noise and can barely be used. For the last binning noise and gain values, please check the CCD Quality Control table for 2018-09-21. In the following shots you have some examples: CCD2 bin 2x2 slow and CCD2 bin 2x2 fast.
Regarding the reading time for the whole 4 CCD mosaic measured in slow and fast readout modes, we measured the following readout times:
bin 1x1 - 48 sec slow mode (28 sec fast mode)
WFC windowing is possible only in chip 4. To set binning window [X1:X2,Y1:Y2] (with X and Y coordinates to cover the necessary region (CCD coordinates of chip #4) around the object, type:
sys> window 4 '[X1:X2,Y1:Y2]'
where (X1,Y1) and (X2,Y2) are the physical corners of the window defined around the object.
to disable this window, type:
sys> window 4 disable
For example, to define a small central 1'x1' window in CCD4, one should use:
sys> window 4 '[934:1114,1960:2140]'
To define a small central 5'x5' window in CCD4, one should use:
sys> window 4 '[569:1479,1595:2505]'
While to define a central 10'x10' window in CCD4, one should use:
sys> window 4 '[168:1988,1190:3010]'
NOTE: Please do not attempt to use the entire CCD4 as a window! This will crash the system, plus there will be no gain in readout time (which is the same as using the entire 4 chip camera).
NOTE: By defining a central window, the overscan information is lost. In order to keep the overscan, one should define the window close to the edge of the CCD4 overscan region (BIASSEC='[10:2150,4105:4190]'). Warning: In 2018 some WFC ICS disconnections were reported (ex FR 22920), apparently caused by windowing close to the CCD overscan region. We recommend caution and test during daytime, and eventualy use of central windows (far from the CCD edge).
For example, to define a 1'x1' which includes the overscan, one should use:
sys> window 4 '[934:1114,3830:4190]'
(which include in Y both the 180 pixel window and the original overscan from Y=4105 to 4190).
Once again, CCD #4 is the only chip that can be windowed.
Moreover, one can define four different windows at the same time on CCD #4 (provided they do not overlap), namely window 1, window 2, window 3 and window 4. Remember, these are not related with the four chips.
Regarding reading time for different size windows, we measured the following times in slow mode:
Whole CCD4 (2154x4200 pix) - 46 sec (thus, no gain from whole WFC)
|Top | Back|