Previous: All observations
Up: Change checklist
Previous Page: All observations
Next Page: Installation of Cameras
- A nitrogen flush to the etalon cavity is required, with the
flow rate independently controllable via a flow meter.
- The CS100 and the switchbox are mounted on the side of TAURUS itself.
There are 6 cables between the CS100 and the switchbox,
plus a mains supply for each.
- If the IPCS is being used, then in order to implement the handshake
between TAURUS and IPCS, a connection is required from the TAURUS 4MS
socket J110 to the socket on the IPCS CCD controller labelled 'TAURUS SKCC'