changing IDS focus between the 235 and 500 cameras
Bring the ICS and uDAS (pink and orange windows)
down completely using the
Switch off the detector SDSU controller
currently in use from its power supply.
Swap the cables from power supply, the
2 x fibre cables (marked to where they go) and the camera shutter
Switch on the detector SDSU controller
at power supply.
in BOTH the DAS and CI windows at the observing console to configure
for the correct detector.
(235 = EEV10, 500 = TEK5)
These are currently options 10 / 11.
in the DAS acquisition window first and then in the CI window.
to change the camera config in the header data.
This file: http://www.ing.iac.es/~eng/ops/int/ids-camera_change.html ejm