Up: Reference section
Previous Page: CS100_RESET
Next Page: CS100_TABLE
Steps the etalon to a specified position in the 4MS lookup table
Format: CS100_STEP <offset>
Range: 1 < <position> < No. of positions in table
Examples : CS100_STEP 10
Comments: This command is used to step the etalon by applying an appropriate Z offset. The Z-offset to be applied is derived from a lookup table held in the 4MS, which must have been previously calculated using the CS100_TABLE command. Issuing CS100_STEP will only result in a change of state of the etalon if the CS100 has been switched to the etalon (command ETALON_VOLTAGE), external control has been enabled (command CS100_EXT) and the servo loop closed (command CS100_CLOSE).