The typical configuration is CCD3=AUX, CCD6=PFIP.
This may change; DET_SHOW_CONF shows the actual configuration.
Check when CCDs were last filled with liquid N and fill if necessary. Check CCD temperatures are correct (see MIMIC screen for CCD1, etc).
SETUP PFIP(AUX)to setup the default formats:
1280 in X, 1180 in Y for EEV3 & EEV6;
1124 1124 for Tek
WINDOW PFIP(AUX) to window, just answer the questions. [3mm] DISABLE_WINDOWS PFIP(AUX) to disable windows (and ENABLE_WINDOWS to restore). [1mm] SEND CCDn OBEY CANCEL_WINDOWS MOVE h to clear windows completely (h=head number) [2mm] <speed> PFIP(AUX) to change CCD readout speed:
SLOW, STANDARD, QUICK, TURBO, NONASTRO.
TURBO, NONASTRO not recommended for science exposures.
[3mm] BIN PFIP(AUX) set on-chip binning factors in X and Y [5mm]