Previous: Hartmann test using the DMS
Up: Setting up the IPCS
Next: Minimum-FWHM using the VAX
Previous Page: Hartmann test using the DMS
Next Page: Minimum-FWHM using the VAX
Using the command EI obtain 3 or 5 pairs of Hartmann left and right arc-lamp exposures, and copy them onto a scratch area on the Vaxcluster as files HART_Li.DST and HART_Ri.DST where i takes values e.g. 1 - 3 or 1 - 5. Type FIGARO, then FOCUS_HART and respond to the prompts.
The parameters which may be changed by the user are: the number of left-right pairs of images; the wavelength direction (X or Y); the first and last pixels in the orthogonal direction to be used; the start collimator position; the increment in collimator position; and whether the maximum pixel-value is to be determined for each of the images (so that the user can check for saturation).
The cross-correlation between spectra extracted from left and right images is performed by the FIGARO routine SCROSS. In-focus position is determined in two ways: by linear interpolation between two values flanking zero line-shift, and by fitting a least-squares straight line to the relation between focus position and line shift. The program plots this relation on a device of the user's choosing e.g. PERICOM_MG, IKON_1, CANON_P, VAXstation TEK_4010 window.